Analog Output K-Type Thermocouple Amplifier - AD8495 Breakout
Thermocouples are very sensitive, requiring a good amplifier with a cold-compensation reference. We have a couple digital thermocouple amplifiers in the shop already from Maxim. Now we're happy to introduce an excellent analog-output amplifier. This is a very simple sensor to use, and if your microcontroller has analog input capability, you'll be ready to go really fast!
The AD8495 K-type thermocouple amplifier from Analog Devices is so easy to use, we documented the whole thing on the back of the tiny PCB. Power the board with 3-18VDC and measure the output voltage on the OUT pin. You can easily convert the voltage to temperature with the following equation: Temperature = (Vout - 1.25) / 0.005 V. So for example, if the voltage is 1.5VDC, the temperature is (1.5 - 1.25) / 0.005 = 50°C
Each order comes with a 2 pin terminal block (for connecting to the thermocouple), a fully assembled PCB with the AD8495 + TLVH431 1.25V precision voltage reference, and pin header (to plug into any breadboard or perfboard). Goes great with our 1m K-type thermocouple (not included). Not for use with any other kind of thermocouple, K type only!
- Works with any K type thermocouple
- Will not work with any other kind of thermocouple other than K type
- Easy to use analog output
- Temp range with 5V power: -250°C to +750°C output (0 to 5VDC) as long as the thermocouple can handle that range
- Temp range with 3.3V power: -250°C to +410°C output (0 to 3.3VDC) as long as the thermocouple can handle that range
Note: The terminal blocks included with your product may be blue or black.
Learn+The AD8495 K-type thermocouple amplifier from Analog Devices is so easy to use, we documented the whole thing on the back of the tiny PCB.
MAY WE ALSO SUGGEST...
MAY WE ALSO SUGGEST...-
See our Distributors page for a complete list of distributors.